CAUTION: Refrain from excessively removing and installing a cable, as this will lead to premature wear of the o-ring and enable water and dust ingress.
NOTE: Probe cable swap is only available on the Butterfly iQ+/iQ+ Vet. Attempting to detach the cable on the Butterfly iQ/iQ Vet is not supported and may result to permanent damage to the device.
The cable on the Butterfly iQ+/iQ+ Vet probe can be replaced in the case of damage, or when a mobile device with a different connector type needs to be used.
To replace the Butterfly iQ+ cable:
1. Remove the existing cable from the Butterfly iQ+/iQ+ Vet probe. Wrap the probe cable around your palm twice while holding the probe firmly in the other hand. Pull the two apart. Do not use tools to grab the strain relief or the cord, as doing so might damage the cable.
2. Align the connector and the probe prior to installation using the diagram below.
3. Push the cable firmly into the probe body. When the cable is fully installed, it should be expected that there will be a small gap between the strain relief and the probe body.